The Mahoney brand CO2/YAG Laser Power Meter Probe 500 watts for CNC Laser Cutting Machines (10.6 down to 1.06 microns).
The probe can easily pay for itself with just one or several uses.
(1) Receive in-warranty replacement of bad glass CO2 laser tubes from Chinese laser machine manufacturer.
(2) Note laser tube degradation over time to adjust laser machine power settings to ensure laserable materials are not ruined.
The probe measures laser power for YAG (900-1250 nm) and CO2 (8-11 microns). Works for laser technologies including CO2 lasers, fiber lasers, YAG lamp lasers, diode end pumped lasers, and diode side pumped lasers.
Each probe is fully tested, calibrated, and each probe has an engraved serial number.
Comes with instructions and convenient storage container.
ONE YEAR FULL WARRANTY FOR MANUFACTURER DEFECTS.
Mahoney CO2/Fiber Laser Power Meter 0-500 watts
A useful asset.
Indepth product information, videos, and more.
Click for Video: Mahoney Laser Power Meter Probe Locating a Bad Mirror on a CO2 Laser Engraving/Cutting Machine. The Mahoney laser power meter probe is utilized to locate bad optics. Maximum CO2 Laser output power is measured before and after each mirror. In this video, the final mirror is measured and found to be bad. The mirror absorbs roughly 9 percent of the laser energy. In general, mirrors are expected to absorb less than 3%.
Click for Video: How my Mahoney 200 watt CO2 Laser Power Probe saved me thousands and stopped a fake seller. The video provides a sample demonstration of the Mahoney laser power probe, and describes how it saved me from a fake seller of CO2 laser tubes.
Click for Instructable: How to measure CO2 laser power on a Laser Engraving Machine and Laser Cutting Machine using a simplified (low cost) CO2 laser power meter. Includes an online instruction set, or a downloadable pdf for Instructable members.
Click for Instructable: How to test and set a Chinese (China) made CO2 Laser Power Supply to get long life from your CO2 Laser Tube for your Laser Engraving and Laser Cutting Machine System. Includes an online instruction set, or a downloadable pdf for Instructable members.
The market leaders implement measured laser power for material settings. Must know laser power to implement material settings effectively:
(1) Universal Laser Systems referenced materials: Click for document: Materials Settings for ULS M360 and V460. Shows table with laser output power as first setting.
(2) Epilog Laser. Click for document Epilog Laser Fusion Materials Settings. Table contains columns showing laser output power.
(3) Trotec Laser. Click for Rowmark settings for measured 60 watt output Trotec Laser.
(4) Laserbits for all major brands. Click for webpage to Laserbits Laser Settings promoted for Laser University. Requires knowledge of laser output power.
This product is great for owners of laser engraving machines, laser engraving systems, laser cutting machines, laser cutting systems.
Also terrific for owners of CO2 glass tube lasers, CO2 and YAG laser markers, laser marking machines, fiber laser marking machines.
It even works with RF lasers manufactured by Synrad, Coherent, Rofin, Universal Laser, Epilog Laser, Universal Laser Systems, Gravograph, Pinnacle. Also, CO2 glass tube lasers including RECI, EFR, Jilin Yong Li, Mactron Technology, North Star Electron, Shanghai Jinke Intelligent Technology, YL Lasers, Prima, Trumpf, PRC laser.
It works with laser machines made by Rabbit Laser, GW Weike, Yueming, Hans Laser, Beijing New Power, Chutian, Foshan Beyond, Grand Bird, Jinan XYZ, Perfect laser, Suke, and many many more.
Get your quick laser power output reading today and exercise your warranty as needed.
Ensure the best quality for the laser cutting and engraving now.
Mahoney CO2 Laser Power Probe Customer Reviews
5.0 out of 5 stars. A good CO2 power meter. By A Y Van Heugten on July 14, 2014. Very reliable. Less complex than a digital device, but does the job well. You have to wait a few minutes between readings, but that's a small price to pay for the the considerably lower cost of this item compared to digital meters.
5.0 out of 5 stars. Does exactly what is supposed to. By danielvwnwieringen on July 7, 2014. Does exactly what is supposed to. A great way to accurately line up the mirrors. Shipped and Received quickly.
4.0 out of 5 stars. Differential temperature testing. By PatrickK on October 20, 2015. Worked when used as directed, slightly tricky getting back to ambient temperature if using more than once using the "Cold water" technique as took a while to settle at new temperature. Adjustment quite stiff but overall gave good results.
5.0 out of 5 stars. Must buy for novice user. By P. Minke on July 30, 2014. Simple, accurate and a must have for setting up and understanding the operation of CO2 laser. Especially if you bought a cheap Chinese model.
5.0 out of 5 stars. Works great! Easy to use. By James Graham on April 15, 2015. Works great! Easy to use! Bought this to check out my 10 year old Epilog Helix 45, still running at 42 after 10 yrs.
5.0 out of 5 stars. Five Stars. By Richard G. on September 24, 2015. Great product as advertised.
5.0 out of 5 stars. Five Stars. By Daniel ROBERT on November 24, 2015. Very good for the prise.
5.0 out of 5 stars. Five Stars. By Randy Berry on September 24, 2014. Excellent tool.
Customers who viewed this item, also viewed